- Research article
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
Impact of naturally occurring amino acid variations on the detection of HIV-1 p24 in diagnostic antigen tests
© Vetter et al. 2015
Received: 6 January 2015
Accepted: 5 October 2015
Published: 29 October 2015
The detection of HIV-1 p24 antigen in diagnostic tests relies on antibodies binding to conserved areas of the protein to cover the full range of HIV-1 subtypes. Using a panel of 43 different virus-like particles (VLPs) expressing Gag from clinical HIV-1 isolates, we previously found that some highly sensitive tests completely failed to detect p24 of certain VLPs, seemingly unrelated to their subtype. Here we aimed to investigate the reason for this failure, hypothesising that it might be due to single amino acid variations in conserved epitopes.
Using amino acid alignment, we identified single amino acid variations at position 16 or 170 of p24, unique to those VLPs that failed to be detected in certain diagnostic tests. Through DNA-mutagenesis, these amino acids were changed to ones more commonly found at these positions. The impact of these changes on p24 detection was tested in commercial diagnostic tests as well as by Western Blot and ELISA, using epitope-specific antibodies.
Results and Conclusions
Changing positions 16 or 170 to consensus amino acids restored the detection of p24 by the investigated diagnostic tests as well as by epitope-specific antibodies in Western Blot and ELISA. Hence, single amino acid changes in conserved epitopes can lead to the failure of p24 detection and thus to false-negative results. To optimise HIV diagnostic tests, they should also be evaluated using isolates which harbour less-frequent epitope variants.
In HIV diagnostics, immunoassays for the detection of viral capsid protein p24 are employed at the stage of screening and diagnosis . In the so-called 4th generation combination screening tests, p24 is used to identify patients with primary infection, prior to the development of anti-HIV antibodies (seroconversion). Technically, the viral antigen is detected in a sandwich format, employing antibodies binding to distinct regions of the protein for capture and detection. These antibodies should bind to highly conserved regions of the target protein, thus minimizing the risk of reduced antigen sensitivity due to subtype-dependent sequence variability or evolutionary escape, both potentially leading to false-negative results. Even single amino acid variations can suffice to decrease the analytical sensitivity of diagnostic immunoassays, as demonstrated for vaccine-induced variants of hepatitis B surface antigen [2, 3]. In the case of HIV, no such diagnostic escape variants have been described for p24, most likely because they simply go unnoticed: An acute primary infection would be false-negative in a 4th generation screening test and thus not detected unless an HIV RNA nucleic acid test was performed in parallel, and a chronic infection would be detected via a patient’s antibody response.
Manufacturers of diagnostic tests evaluate sequence-associated test sensitivity by testing HIV strains from different subtypes and frequent circulating recombinant forms (CRFs), thus hoping to cover a range of sequence variants. However, in a recent study conducted by our laboratory, using 43 virus-like-particles (VLPs) derived from clinical HIV-1 isolates expressing the Gag proteins of different subtypes and CRFs, we noticed that the failure of a diagnostic test to detect p24 of certain virus isolates was not necessarily related to their subtype, as even otherwise highly sensitive tests failed to detect a common subtype B VLP . We hypothesised that detection failure could be due to subtype-independent amino acid variations and analysed our p24 sequences in relation to amino acid variation and detection failure. In this study, we sought to identify amino acid variations that abrogated the detection of p24 in diagnostic HIV-1 p24 antigen-only or 4th-generation HIV screening assays, or by epitope-specific antibodies used in home-made tests.
Virus-like particle cloning, production and quantification
VLPs used in this study were expressed from Gag-Pol encoding eukaryotic expression vectors, which were cloned for a previous study conducted in our laboratory . Briefly, the gag-pr region of patient-derived viral RNA was cloned into the VLP backbone vector of pCMVΔ8.91, after removing its existing gag-pr portion. VLPs were produced by transiently transfecting 293 T cells and VLPs were purified via ultracentrifugation over a 20 % sucrose cushion. VLP quantification and normalisation of concentration was based on reverse-transcriptase (RT) activity (Roche Colorimetric Reverse Transcriptase Assay, article 11468120910) which was put into relation to p24 content and international units based on the World Health Organization p24 reference standard (90/636). One IU/ml of this standards equals approximately 5.1 pg/ml of p24 (bioMérieux VIDAS HIV p24 II) and 0.0005 ng/ml of RT (Roche RT assay) (for more details see ).
The Los Alamos National Laboratory database (http://www.hiv.lanl.gov/content/sequence/HIV/mainpage.html) was queried for all available HIV-1 p24 sequences (of any subtype). One sequence per patient (option in LANL database) was downloaded in aligned amino acid table format and the frequency analysis was conducted in Excel.
Amino acid changes at position 16 and 170 in p24 were introduced by side-directed-mutagenesis of the VLP eukaryotic expression vectors, using the QuickChange II XL Site-Directed Mutagenesis Kit (200521, Agilent Technologies). Primers were as follows: pBV8-B-T16S 5’-GTACATCAGGCCCTATCACCTAGAACTTTAAATG-3’; pBV59-F2-T16S 5’-GTACATCAGCCTCTATCACCTAGAACTTTAAATG-3’ ; pBV11-B-S16T 5’-GTACATCAGGCCATAACACCTAGAACTTTAAATG-3’; pBV43-D-R170K 5’-GATTATGTAGATCGGTTCTATAAAACTCTAAGAGCCGAGC-3’; pBV60-D-R170K 5’-GATCGGTTCTATAAAACTCTAAGAGCC-3’; pBV42-12BF-N170K 5’-GACAGGTTCTTTAAGACCCTAAGAGCC-3’; pBV38-D-K170R 5’-GACTATGTAGATCGGTTCTATAGAACTCTAAGAGCCGAGC-3’. Mutagenesis was verified by sequencing (for sequencing primers see ).
VLPs were diluted to 10 IU/ml p24 antigen (or 0.005 ng/ml RT) in negative human plasma (Swiss Red Cross Blood Donation Centre Zurich) in a single batch per VLP and stored at −20 °C. Antigen reactivity of the (anti-HIV-negative) samples was analysed with the following CE-marked and FDA-approved diagnostic tests and platforms: Abbott Architect HIV Ag/Ab Combo; bioMérieux VIDAS HIV p24 II; Siemens Enzygnost Integral II; Siemens Enzygnost Integral 4.
VLPs in PBS were diluted in NuPage LDS sample buffer and reducing agent (NP0007 and NP0009, Life Technologies) to a final concentration of 10 ng/ml RT. Samples were heated for 10 min at 95 °C. SDS polyacrylamide gel electrophoresis was performed using 10 % gels in standard Tris/Glycine buffer, and proteins were blotted on nitrocellulose membrane by wet transfer over night. Primary mouse monoclonal antibodies used to detect HIV p24 were AG3.0 (4121, NIH AIDS Reagent Program) and YDHIV1gp24 (sc-73300, Santa Cruz). Polyclonal human anti-p24 plasma was obtained by pooling plasma from 10 HIV-positive individuals, inactivated with 0.5 % Triton X-100. Species-specific HRP-coupled secondary antibodies were obtained from KPL (goat-anti-mouse HRP 474–1806; goat-anti-human HRP 141–992). Super Signal West Pico (34077, Thermo Scientific) was used as chemiluminescent substrate and the reaction visualised in the ImageQuant LAS-4000 imager (GE Healthcare).
VLPs in PBS were lysed in 3x SNCR virus disruption buffer  and coated onto clear 384-well plates (Nunc MaxiSorb P6366, Sigma Aldrich) at a concentration of 5 ng/ml RT over night at 4 °C. After blocking with 2 % non-fat milk in PBS, primary and secondary antibodies (see Western Blot) were sequentially applied for 1 h with intermittent washes. TMB liquid substrate (T8665, Sigma Aldrich) was diluted at 1:2 in H2O and added for 15 min, followed by the addition of 1 M H2SO4 to stop the reaction. Optical density was measured at 450 nm on the Perkin Elmer EnVision microplate reader. Cut-off values were derived from wells mock-coated with the same reagent, but lacking antigen, and processed in the same way.
This study did not use any personal information, but only previously published viral sequences . Therefore, in full accordance with Swiss legislation (Federal Act on Data Protection of 19 June 1992; see https://www.admin.ch/opc/en/classified-compilation/19920153/index.html) informed consent was not necessary.
Amino acid variants
To identify amino acid variations which might be responsible for the failure to detect p24 in a diagnostic test, the p24 sequences of all 43 VLP panel members investigated in our previous study were aligned (sequences available at NCBI GenBank accession numbers KJ689249-KJ689290) . This highlighted a single amino acid variation of Serine (S) to Threonine (T) at position 16 of pBV8-B and pBV59-F2; both VLPs had been the only ones poorly detected in two highly sensitive tests in our previous study (Siemens Enzygnost Integral 4 and Innogenetics Innotest HIV Antigen mAb). Furthermore, we identified a variation of Lysine (K) to Arginine (R) or Asparagine (N) at position 170, which might be accountable for the failure of the bioMérieux VIDAS DuoUltra and VIDAS HIV p24 II to detect VLPs pBV43-D, pBV60-D and pBV42-12BF. To ensure that amino acid variations in the Gag VLPs were not erroneously introduced during the Gag cloning procedure , the relevant part of the original viral RNA isolated from patient plasma was sequenced, and the amino acid variations at position 16 and 170 of the respective VLPs were confirmed at nucleotide level in the viral genome (data not shown).
Frequency of amino acid variations for positions 16 and 170 of p24 of sequences deposited in the Los Alamos National Laboratory sequence database
Amino acids at position 16
Amino acids at position 170
Other group M
Summary of VLPs investigated in this study
AA position in p24 (Gag)a
AA in wt VLP
AA change introduced
HIV diagnostic test with low sensitivity for wt
Siemens Enzygnost Integral II/4
Siemens Enzygnost Integral II/4
N/A – control VLP
bioMérieux VIDAS HIVp24 II
bioMérieux VIDAS HIVp24 II
bioMérieux VIDAS HIVp24 II
N/A – control VLP
Similarly, analysis of the position 170 wildtype and mutant VLPs on the bioMérieux VIDAS p24 II (Fig. 1d) showed a restored ability to detect the mutant pBV43-D-R170K, pBV60-D-R170K and pBV42-12BF-N170K. Introducing the amino acid variant (R) at position 170 into the wildtype pBV38-D, reduced, but not completely abolished, detection by the bioMérieux VIDAS HIV p24 II. The somewhat lower re-gained detectability of pBV43-D-R170K and pBV60-D-R170K and the only two-fold (rather than complete) loss off pBV38-D-K170R signal suggest that additional, unidentified epitope variations in these VLPs also play a role in detection by this test. Again, all VLPs were also tested on the Abbott Architect to control for the presence of viral antigen (panel E). These observations demonstrate that a single amino acid variation can account for the complete failure of p24 antigen detection by certain diagnostic HIV-1 tests.
Western blot and ELISA
Modern HIV 4th generation combination screening tests or antigen-only tests employed for HIV screening and diagnosis, are expected to detect all HIV-1 subtypes and circulating recombinant forms. Many of these tests currently on the market indeed have a fairly good sensitivity and subtype breadth [4, 8]. We were thus surprised that some otherwise highly sensitive tests failed to detect certain VLPs at a p24 concentration of 10 IU/ml, which roughly corresponds to 500’000 cp/ml RNA (for conversion see ). The analyses conducted in the present study suggest that this failure is not necessarily due to a set of subtype-specific amino acid sequence variations, but rather due to single amino acid variations. The amino acids at position 16 and 170 identified here as being responsible for detection failure, are both located in highly conserved regions of p24 and deviations from the consensus sequence are of conservative nature. Although position 16 is part of the well-studied HLA-B*57-restricted CTL epitope ISW9  and hence target of selective pressure, escape variants have only been described for positions 14 and 15 [10, 11]. It’s role might be rather of structural nature, as position 16 is part of an N-terminal three amino acid linker-region (14–16) in p24 , which plays a role in protein re-folding upon proteolytic processing of Gag . Position 170 is part of the major homology region (MHR) of p24 (153–172), the most conserved part of the viral capsid found in many retroviruses, with an essential role in Gag assembly [14, 15] and viral infectivity . The rather conservative nature of the K to R variant and the relative low frequency of less conservative substitutions at position 170 are in line with the importance of conserving the MHR function. An Alanine exchange at K170 was found to maintain assembly function but not infectivity of viral particles [16, 17]. Notably, a certain degree of flexibility seems to exist, given the K to N change (relatively large positive to smaller uncharged properties) observed in pBV42-12BF. It is unknown if this requires compensatory second-site changes in the protein.
The frequency of S16T and K170R variations found in the LANL deposited sequences, suggests that as many as 1:50–1:60 patients carry these variations. While it is not known how frequent they are in primary infection, their importance in the use of immunoassays should be kept in mind when selecting epitope regions for generation of antibodies for diagnostic tests. Our previous study suggests that aside from the Siemens Enzygnost Integral II and 4, the Innotest HIV Antigen mAb from Innogenetics also targets position 16. Here too, pBV8-B and pBV59-F2 were the only VLPs of the whole panel to be poorly detected . Furthermore, in addition to the bioMérieux 4th generation combo test (DuoULTRA) and antigen-only test (HIV p24 II), the Siemens ADVIA CHIV also had poor sensitivity for pBV43-D, pBV60-D and pBV42-12BF, indicating binding of antigen around position 170 in this test. Notably, the bioMérieux tests detected several VLPs with severely lower sensitivity , but obvious amino acid variations could only be identified for the three VLPs carrying the position 170 amino acid variant. This suggests that other epitopes targeted by this test are also sensitive to sequence variation.
Although both amino acid variations described in this study are of rather conservative nature, their impact on antibody binding and detection of p24 by diagnostic HIV tests is striking. It demonstrates that test sensitivity and breadth of detection are not only defined by subtypes and their associated overall sequence variations, but also by single amino acid polymorphisms. Future antigen test development should take these findings into account to cover the broadest spectrum of circulating isolates possible.
We would like to thank the staff of all participating laboratories for their help in this study.
The work of BNV and JS was financed partially by contract 11.00526/304.0001-697 with the Swiss Federal Office of Public Health. The funder had no role in the design, conduct or interpretation of this study.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Daskalakis D. HIV diagnostic testing: evolving technology and testing strategies. Topics Antiviral Med. 2011;19(1):18–22.Google Scholar
- Huang X, Qin Y, Li W, Shi Q, Xue Y, Li J, et al. Molecular analysis of the hepatitis B virus presurface and surface gene in patients from eastern China with occult hepatitis B. J Med Virol. 2013;85(6):979–86.View ArticlePubMedGoogle Scholar
- Huang CH, Yuan Q, Chen PJ, Zhang YL, Chen CR, Zheng QB, et al. Influence of mutations in hepatitis B virus surface protein on viral antigenicity and phenotype in occult HBV strains from blood donors. J Hepatol. 2012;57(4):720–9.View ArticlePubMedGoogle Scholar
- Vetter BN, Orlowski V, Fransen K, Niederhauser C, Aubert V, Brandenberger M, et al. Generation of a recombinant Gag virus-like-particle panel for the evaluation of p24 antigen detection by diagnostic HIV tests. PLoS One. 2014;9(10):e111552.View ArticlePubMedPubMed CentralGoogle Scholar
- Schupbach J, Tomasik Z, Knuchel M, Opravil M, Gunthard HF, Nadal D, et al. Optimized virus disruption improves detection of HIV-1 p24 in particles and uncovers a p24 reactivity in patients with undetectable HIV-1 RNA under long-term HAART. J Med Virol. 2006;78(8):1003–10.View ArticlePubMedGoogle Scholar
- Sanders-Beer BE, Eschricht M, Seifried J, Hirsch VM, Allan JS, Norley S. Characterization of a monoclonal anti-capsid antibody that cross-reacts with three major primate lentivirus lineages. Virology. 2012;422(2):402–12.View ArticlePubMedGoogle Scholar
- Tang S, Zhao J, Wang A, Viswanath R, Harma H, Little RF, et al. Characterization of immune responses to capsid protein p24 of human immunodeficiency virus type 1 and implications for detection. Clin Vaccine Immunol: CVI. 2010;17(8):1244–51.View ArticlePubMedPubMed CentralGoogle Scholar
- Ly TD, Plantier JC, Leballais L, Gonzalo S, Lemee V, Laperche S. The variable sensitivity of HIV Ag/Ab combination assays in the detection of p24Ag according to genotype could compromise the diagnosis of early HIV infection. J Clin Virol. 2012;55(2):121–7.View ArticlePubMedGoogle Scholar
- Goulder PJ, Bunce M, Krausa P, McIntyre K, Crowley S, Morgan B, et al. Novel, cross-restricted, conserved, and immunodominant cytotoxic T lymphocyte epitopes in slow progressors in HIV type 1 infection. AIDS Res Hum Retroviruses. 1996;12(18):1691–8.View ArticlePubMedGoogle Scholar
- Draenert R, Le Gall S, Pfafferott KJ, Leslie AJ, Chetty P, Brander C, et al. Immune selection for altered antigen processing leads to cytotoxic T lymphocyte escape in chronic HIV-1 infection. J Experimental Med. 2004;199(7):905–15.View ArticleGoogle Scholar
- Crawford H, Lumm W, Leslie A, Schaefer M, Boeras D, Prado JG, et al. Evolution of HLA-B*5703 HIV-1 escape mutations in HLA-B*5703-positive individuals and their transmission recipients. J Experimental Med. 2009;206(4):909–21.View ArticleGoogle Scholar
- Gitti RK, Lee BM, Walker J, Summers MF, Yoo S, Sundquist WI. Structure of the amino-terminal core domain of the HIV-1 capsid protein. Science. 1996;273(5272):231–5.View ArticlePubMedGoogle Scholar
- von Schwedler UK, Stemmler TL, Klishko VY, Li S, Albertine KH, Davis DR, et al. Proteolytic refolding of the HIV-1 capsid protein amino-terminus facilitates viral core assembly. EMBO J. 1998;17(6):1555–68.View ArticleGoogle Scholar
- Schur FK, Hagen WJ, Rumlova M, Ruml T, Muller B, Krausslich HG, et al. Structure of the immature HIV-1 capsid in intact virus particles at 8.8 A resolution. Nature 2015;517(7535):505–8.View ArticlePubMedGoogle Scholar
- Mammano F, Ohagen A, Hoglund S, Gottlinger HG. Role of the major homology region of human immunodeficiency virus type 1 in virion morphogenesis. J Virol. 1994;68(8):4927–36.PubMedPubMed CentralGoogle Scholar
- von Schwedler UK, Stray KM, Garrus JE, Sundquist WI. Functional surfaces of the human immunodeficiency virus type 1 capsid protein. J Virol. 2003;77(9):5439–50.View ArticleGoogle Scholar
- Chang YF, Wang SM, Huang KJ, Wang CT. Mutations in capsid major homology region affect assembly and membrane affinity of HIV-1 Gag. J Mol Biol. 2007;370(3):585–97.View ArticlePubMedGoogle Scholar