- Research article
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
The development of a GeXP-based multiplex reverse transcription-PCR assay for simultaneous detection of sixteen human respiratory virus types/subtypes
- Jin Li†1,
- Nai-Ying Mao†1,
- Chen Zhang1,
- Meng-Jie Yang1,
- Miao Wang1,
- Wen-Bo Xu1Email author and
- Xue-Jun Ma1Email author
© Li et al.; licensee BioMed Central Ltd. 2012
Received: 13 February 2012
Accepted: 10 August 2012
Published: 14 August 2012
Existing standard non-molecular diagnostic methods such as viral culture and immunofluorescent (DFA) are time-consuming, labor intensive or limited sensitivity. Several multiplex molecular assays are costly. Therefore, there is a need for the development of a rapid and sensitive diagnosis of respiratory viral pathogens.
A GeXP-based multiplex RT-PCR assay (GeXP assay) was developed to detect simultaneously sixteen different respiratory virus types/subtypes. Seventeen sets of chimeric primers were used to initiate the RT-PCR, and one pair of universal primers was used for the subsequent cycles of the RT-PCR. The specificity of the GeXP assay was examined with positive controls for each virus type/subtype. The sensitivity was evaluated by performing the assay on serial ten-fold dilutions of in vitro-transcribed RNA of all RNA viruses and the plasmids containing the Adv and HBoV target sequence. GeXP assay was further evaluated using 126 clinical specimens and compared with Luminex xTAG RVP Fast assay.
The GeXP assay achieved a sensitivity of 20–200 copies for a single virus and 1000 copies when all of the 16 pre-mixed viral targets were present. Analyses of 126 clinical specimens using the GeXP assay demonstrated that GeXP assay and the RVP Fast assay were in complete agreement for 109/126 (88.51%) of the specimens. GeXP assay was more sensitive than the RVP Fast assay for the detection of HRV and PIV3, and slightly less sensitive for the detection of HMPV, Adv, RSVB and HBoV. The whole process of the GeXP assay for the detection of 12 samples was completed within 2.5 hours.
In conclusion, the GeXP assay is a rapid, cost-effective, sensitive, specific and high throughput method for the detection of respiratory virus infections.
Viral respiratory tract infections, which have a considerable morbidity and fatality rate, are common diseases that especially affect infants and the elderly . Common respiratory viruses include influenza A virus, influenza B virus, parainfluenza virus, human rhinovirus, adenovirus and respiratory syncytial virus. New respiratory viruses important to public health, such as metapneumovirus, coronavirus (subtypes SARS-Cov and CoV HKU1) and human bocavirus [2, 3], have emerged over the past decade. The clinical presentation of respiratory infections caused by different viral pathogens can be very similar, making etiological diagnosis difficult .
The traditional assays used to diagnose respiratory tract viruses are viral culture and immunofluorescent staining. Viral culture remains the gold standard for the diagnosis of respiratory viruses because of its broad spectrum and high specificity. However, viral culture is time-consuming and has a low sensitivity for the detection of some respiratory viruses that have fastidious growth requirements [5, 6]. Immunofluorescent staining is fast, but the assay sensitivity and the availability of antisera can be limiting factors [5, 6]. With the development of rapid molecular techniques, molecular assays, especially in a multiplex format, have been accepted as tests of choice for broad spectrum detection of respiratory viruses. Several multiplex assays are available commercially such as xTAG RVP from Luminex [7, 8], Multicode-PLx RVP from EraGen Biosciences , and ResPlex II from Qiagen . However, all of them are based on a liquid-phase bead-based array technology to implement the detection, which has increased the cost and the implementation time of the whole assays.
The GenomeLab Gene Expression Profiler genetic analysis system (GeXP) developed by Beckman Coulter (Brea, CA, USA) is a new multitarget, high-throughput detection platform that integrates reverse transcription-PCR (RT-PCR) and labeled amplified products in a multiplex PCR assay, followed by fluorescence capillary electrophoresis separation based on the size of the amplified products. This system has been used previously in identifying rapidly gene expression prostate cancer biomarker signatures in biological samples, rapid and sensitive detection of 68 unique varicella zoster virus gene transcripts  and detection of pandemic influenza A H1N1 virus  and nine serotypes of enteroviruses associated with hand, foot and mouth disease .
In this report, a novel RT-PCR assay using the GeXP (GeXP assay) for rapid, sensitive, multiplex detection of sixteen different respiratory virus types/subtypes: influenza A virus (FluA), influenza B virus (FluB), seasonal influenza A H1N1 virus (sH1N1), parainfluenza virus type 1 (PIV1), parainfluenza virus type 2 (PIV2), parainfluenza virus type 3 (PIV3), human rhinovirus (HRV), human metapneumovirus (HMPV), adenovirus (Adv), respiratory syncytial virus A (RSVA), respiratory syncytial virus B (RSVB), four coronavirus sybtypes (CoV HKU1, CoV NL63, CoV 229E, CoV OC43) and human bocavirus (HBoV) was described. The specificity and sensitivity of the GeXP assay were examined, and the clinical performance of the GeXP assay was evaluated by comparing the results obtained by the GeXP assay to those obtained by the Luminex xTAG RVP Fast kit (Luminex Corporation, Toronto, Canada) for 126 nasopharyngeal aspirates collected from hospitalized children.
Viruses and controls
Cell culture virus stocks from the NATtrolTM Respiratory Validation Panel 2 (NATRVP-2) (ZeptoMetrix, New York, USA) were used as the positive controls for FluA H3, FluB, sH1N1, PIV1-3, HRV, Adv 3, RSVA, RSVB, HMPV, CoV 229E and CoV OC43. Clinical specimens that were genotyped and sequenced previously by the Biotech Center for Viral Disease Emergency, a part of the Chinese Center for Disease Control (CCDC), were used as positive controls for HBoV, CoV HKU1 and CoV NL63.
To evaluate the sensitivity of the assay, controls were prepared with dilutions from 10 to 105 copies of vectors containing PCR products cloned from each virus individually. The PCR products were cloned into a pGEM-T vector, which was used to transform DH10B cells. The plasmid DNA was extracted with an E.Z.N.A. Plasmid Mini Kit I (Omega, GA, USA). The RNA copy number was calculated after measuring the concentration of the RNA transcribed in vitro using a T7 Large Scale RNA Production System (Promega, Wisconsin, USA). For the DNA viruses Adv and HBoV, the in vitro transcription was omitted.
A total of 126 nasopharyngeal aspirates were collected from children under two years of age who were hospitalized at the Children’s Hospital of Beijing, China, during June, 2008 and March, 2010 with a diagnosis of pneumonitis or bronchopneumonia and a fever of 38°C or greater. A total volume of 0.5 ml of nasopharyngeal aspirate was collected in 3.5 ml of transport medium. This study was approved by the Beijing ethics committee. Sample collection was agreed by child's parents or grandparents with informed consent.
Extraction and purification of RNA/DNA
Total RNA/DNA was extracted from 200 μl of viral stocks or clinical samples using the QIAamp Viral RNA Mini Kit (Qiagen, Hilden, Germany) according to the manufacturer’s instructions. The extracts was eluted in 50 μl of DNase- and RNase-free water and stored at −80°C.
GeXP assay and detection
The multiplex RT-PCR was performed with a One Step RT-PCR kit (Qiagen, Hilden, Germany) in a 25 μl volume, containing 5 μl of the 5× buffer, 1 μl of the dNTP mix, 1 μl of the enzyme mix, 1.25 pmol each of the forward chimeric primer mix and reverse chimeric primer mix, 12.5 pmol each of the forward universal primer mix and reverse universal primer mix, 2 μl of template RNA and 0.1 μl of ribonuclease inhibitor (Takara, Dalian, China) and DNase- and RNase-free water. The RT-PCR mixture was subjected to the following amplification conditions: 50°C for 30 min, 95°C for 15 min, followed by 10 cycles of 95°C for 30 s, 55°C for 30 s, 72°C for 30 s; 10 cycles of 95°C for 30 s, 65°C for 30 s, 72°C for 30 s; 20 cycles of 95°C for 30 s, 48°C for 30 s, 72°C for 30 s, and a final incubation of 72°C for 3 min. Two μl of each Cy5-labeled PCR product was separated via GeXP capillary electrophoresis, and the dye signal strength was measured by fluorescence spectrophotometry in arbitrary units (A.U.) of optical fluorescence. For all amplified products, the reaction was considered positive when the value of dye signal was over 2000 A.U.
Specificity and sensitivity of the GeXP assay
The specificity of the GeXP assay for all viral targets was tested individually in a multiplex assay under the experimental conditions of the GeXP assay. The sensitivity of the GeXP assay for all viral targets was examined individually in a multiplex assay using serial ten-fold dilutions from 10 to 105 copies of in vitro-transcribed RNA of all RNA viruses and the plasmids containing the Adv and HBoV target sequences. The sensitivity of the GeXP assay was also examined by using 16 pre-mixed quantitative viral targets.
RVP fast assay and detection
The Luminex xTAG RVP Fast kit (Luminex Corporation, Toronto, Canada) enables users to detect simultaneously FluA, FluB, RSV, PIV1-4, AdV, HMPV, CoV 229E, NL63, OC43, and HKU1, enterovirus, HRV and HBoV. The RNA/DNA extracted from 126 clinical specimens was tested using the RVP Fast assay in a 96-well plate format according to the manufacturer’s instructions (Luminex Corporation, Toronto, Canada). The plate was then analyzed using the Bio-PlexTM 200 system (Bio-Rad, CA, USA) according to the manufacturer’s instructions, and the median fluorescence intensity (MFI) was determined.
Specificity and sensitivity of the GeXP assay
The sensitivity of the assay was evaluated individually for each virus in the multiplex assay using serial ten-fold dilutions of cloned PCR products. In the multiplex assay, the detectable level of HRV and PIV2 was 20 copies per reaction, while the limit of detection for the other 14 viruses types/subtypes was 200 copies per reaction. The clones with the same copies of each virus were then pre-mixed to evaluate the sensitivity when all of the 16 viral targets were present. The detection sensitivity of all of the pre-mixed viral targets was 1000 copies per reaction in the multiplex assay (Figure 1. R).
Evaluation of the GeXP assay using respiratory specimens
All 126 specimens detected by the RVP Fast assay, the reference method, were retested by the GeXP assay. The results from both assays showed that HRV was found most frequently (68/126), followed by RSVB (46/126) and PIV3 (21/126) in the 126 specimens. A total of 23 negative specimens (18.25%) were detected by the GeXP assay and 24 (19.05%) negative specimens by the RVP Fast assay. One negative specimen detected by the RVP Fast assay was found to be positive for HRV by the GeXP assay. A total of 66 specimens with co-infections were detected by the RVP Fast assay, 64 of them were co-infections detected by the GeXP assay, and 4 additional co-infections were detected only by the GeXP assay and confirmed by sequencing as true positives.
Detection of 16 respiratory viruses in 126 specimens
No. of specimens
Performance of the GeXP assay
GeXP + aRVP + a
GeXP + RVP -
GeXP- RVP +
A one-step, sensitive, specific assay using GeXP for the simultaneous detection of 16 respiratory virus types/subtypes (FluA, FluB, sH1N1,PIV1, PIV2, PIV3, HRV, RSVA, RSVB, HMPV, Adv, CoV OC43, CoV 229E, CoV NL63, CoV HKU1 and HBoV) was described in this study. The GeXP genetic analysis system is a new multitarget, high throughput detection platform, and its application in the differential detection of pandemic Influenza A H1N1 virus and seasonal Influenza A virus was reported recently by our laboratory . To further reduce the assay time and avoid contamination, a one-step multitarget RT-PCR was performed in one tube rather than multiplex two-step RT-PCRs in separate tubes  would be preferred. To develop a one-step GeXP assay, several commercial one-step RT-PCR kits from different companies (Omega, Bio-Rad, Qiagen and Invitrogen) were tested in our preliminary experiments. The one-step RT-PCR kit from Qiagen revealed the best amplification to perform under our current protocol (data not shown).
The temperature switch PCR (TSP)  strategy was adopted to optimize the amplification parameters. The biphasic PCR parameters of the TSP allow a multiplex PCR to be performed under standardized PCR conditions, and therefore do not require optimization of each individual PCR assay. The optimal settings for three different denaturation temperatures and the amplification cycle conditions were determined in the current protocol. The chimeric primers consisted of a specific primer sequence fused to the universal primer sequence. The primers were designed to generate gene fragments with lengths between 140–340 bp. The amount of universal primers included in the RT-PCR was ten times that of the chimeric primers, so in the last 20 cycles of PCR, amplification was carried out predominantly by a single pair of universal primers, of which only the forward universal primer was labeled fluorescently. This should reduce the occurrence of preferential amplification in the reaction. Internal control primers were added to the reaction to ascertain whether the extraction and reverse-transcription steps of the assay were functioning correctly. Only 2 μl of the cy5-labeled PCR amplified fragments were separated by capillary electrophoresis based on size and detected by the GeXP system.
The specificity of the GeXP assay was examined using artificial specimens from the NATRVP-2 panel and clinical specimens confirmed previously to be positive. A specific peak of PCR product was obtained only for the expected viral target using the GeXP system. No cross-reactivity among the 16 respiratory virus types/subtypes was observed. The detection sensitivity for each virus was 20–200 copies per reaction when the assay was performed separately for each virus, and the sensitivity was 1000 copies per reaction when all of the 16 pre-mixed viral targets were present in the multiplex assay. The results indicate that the detection sensitivity of the GeXP assay for mixed virus samples was slightly lower than that for a single type of virus. The detection sensitivity for each virus in this study was similar to that of real-time PCR assays reported in recent years [18–20].
Analyses of 126 specimens using the GeXP assay and RVP Fast assay demonstrated that the GeXP assay had comparable sensitivity and specificity to the commercially available RVP Fast assay (Table 2). Discrepancies in detection results were found for HRV, PIV3, HMPV, Adv, RSV B and HBoV. All of the 9 specimens positive for PIV3 or HRV detected only by the GeXP assay were confirmed by independent PCR and sequencing to be true positives (data not shown), suggesting that the GeXP assay is more sensitive than the RVP Fast assay for the detection of HRV and PIV3. Because the HRV primers used in the RVP Fast assay were able to amplify both HRV and enterovirus, the three specimens positive for HRV detected only by the RVP Fast assay could actually be enteroviruses. All of the specimens positive for HRV detected by the GeXP assay were confirmed by sequencing as true HRV positives. Two of HMPV negative samples, 1 of Adv negative sample, 3 of RSVB negative samples and 4 of HBoV negative samples detected by the GeXP assay were positive by the RVP Fast assay. All of these negative samples had lower median florescence intensity (MFI) values (294–825) in the RVP Fast assay, suggesting that the GeXP assay has a slightly decrease sensitivity for the detection of HMPV, Adv, RSVB and HBoV. However, the difference was not significant (for HMPV, Adv, RSVB and HBoV, p = 0.5, 1, 0.25, 0.125, respectively, using McNemar’s test; Kappa = 0.880, 0.943, 0.949, 0.864, respectively). The detection of coronaviruses (CoV HKU1, CoV NL63, CoV 229E and CoV OC43) by both assays was completely consistent between assays. The overall detection rate of the GeXP assay for each virus was comparable to that of the RVP Fast assay, demonstrating the high sensitivity and specificity of the GeXP assay in the analysis of clinical samples. It should be noted that there are not sufficient positive samples for FluA, sH1N1, FluB, PIV1,PIV2, CoV NL63, CoV 229E, CoV HKU1 and RSVA to determine meaningful statistics in our study. Due to the limited specimens available at this time, only preliminary findings were reported in this study. The detection rate for infections involving two or more viruses was similar for the GeXP assay (68/126) and the RVP Fast assay (66/126). Most occurrences of multiplex infections involved two of three viruses, HRV, RSVB and PIV3 (data not shown).
Two distinct advantages of the GeXP assay are the short assay time and the low cost per test. Though the cost of GeXP equipment is approximately $100.000, the cost of the GeXP assay for simultaneous detection of 16 respiratory virus types/subtypes is approximately $8 per test (the RT-PCR kit and the consumables of detection), versus $120 per test using the RVP Fast kit or $8 per test for each virus using a commercial RT-PCR kit (DaAn, Gene, China). The whole reaction was completed in one tube in a one-step multiplex RT-PCR within 2.5 hours, followed by capillary electrophoresis separation (10 min/12 wells). In addition, two 96-well plates can be placed in parallel in a GeXP machine at the same time to further increase the throughput of the samples.
In summary, this study has demonstrated that the GeXP assay is a rapid, cost-effective and high throughput method with high sensitivity and specificity for the detection of respiratory virus infection. Further validation of the GeXP assay with a larger number of clinical samples is necessary before this method can be used widely for routine laboratory testing in China.
We acknowledge the Biotech Center for Viral Disease Emergency, Beijing, China gratefully for providing the clinical specimens used in this study. This work was supported by the China Mega Project for Infectious Disease (2011ZX10004-001, 2009ZX10004-101, 2012ZX10004-215).
- Cant AJ, Gordon SB, Read RC, Hart CA, Winstanley C: Respiratory infections. J Med Microbiol. 2002, 51 (11): 903-914.View ArticlePubMedGoogle Scholar
- Fauci AS, Touchette NA, Folkers GK: Emerging infectious diseases: a 10-year perspective from the National Institute of Allergy and Infectious Diseases. Emerg Infect Dis. 2005, 11 (4): 519-525. 10.3201/eid1104.041167.View ArticlePubMedPubMed CentralGoogle Scholar
- Mahony J, Chong S, Merante F, Yaghoubian S, Sinha T, Lisle C, Janeczko R: Development of a respiratory virus panel test for detection of twenty human respiratory viruses by use of multiplex PCR and a fluid microbead-based assay. J Clin Microbiol. 2007, 45 (9): 2965-2970. 10.1128/JCM.02436-06.View ArticlePubMedPubMed CentralGoogle Scholar
- Coiras MT, Aguilar JC, Garcia ML, Casas I, Perez-Brena P: Simultaneous detection of fourteen respiratory viruses in clinical specimens by two multiplex reverse transcription nested-PCR assays. J Med Virol. 2004, 72 (3): 484-495. 10.1002/jmv.20008.View ArticlePubMedGoogle Scholar
- Henrickson KJ: Advances in the laboratory diagnosis of viral respiratory disease. Pediatr Infect Dis J. 2004, 23 (1 Suppl): S6-S10.View ArticlePubMedGoogle Scholar
- Storch GA: Diagnostic virology. Clin Infect Dis. 2000, 31 (3): 739-751. 10.1086/314015.View ArticlePubMedGoogle Scholar
- Krunic N, Yager TD, Himsworth D, Merante F, Yaghoubian S, Janeczko R: xTAG RVP assay: analytical and clinical performance. J Clin Virol. 2007, 40 (Suppl 1): S39-S46.View ArticlePubMedGoogle Scholar
- Pabbaraju K, Wong S, Tokaryk KL, Fonseca K, Drews SJ: Comparison of the Luminex xTAG respiratory viral panel with xTAG respiratory viral panel fast for diagnosis of respiratory virus infections. J Clin Microbiol. 2011, 49 (5): 1738-1744. 10.1128/JCM.02090-10.View ArticlePubMedPubMed CentralGoogle Scholar
- Nolte FS, Marshall DJ, Rasberry C, Schievelbein S, Banks GG, Storch GA, Arens MQ, Buller RS, Prudent JR: MultiCode-PLx system for multiplexed detection of seventeen respiratory viruses. J Clin Microbiol. 2007, 45 (9): 2779-2786. 10.1128/JCM.00669-07.View ArticlePubMedPubMed CentralGoogle Scholar
- Li H, McCormac MA, Estes RW, Sefers SE, Dare RK, Chappell JD, Erdman DD, Wright PF, Tang YW: Simultaneous detection and high-throughput identification of a panel of RNA viruses causing respiratory tract infections. J Clin Microbiol. 2007, 45 (7): 2105-2109. 10.1128/JCM.00210-07.View ArticlePubMedPubMed CentralGoogle Scholar
- Nagel MA, Gilden D, Shade T, Gao B, Cohrs RJ: Rapid and sensitive detection of 68 unique varicella zoster virus gene transcripts in five multiplex reverse transcription-polymerase chain reactions. J Virol Methods. 2009, 157 (1): 62-68. 10.1016/j.jviromet.2008.11.019.View ArticlePubMedPubMed CentralGoogle Scholar
- Qin M, Wang DY, Huang F, Nie K, Qu M, Wang M, Shu YL, Ma XJ: Detection of pandemic influenza A H1N1 virus by multiplex reverse transcription-PCR with a GeXP analyzer. J Virol Methods. 2010, 168 (1–2): 255-258.View ArticlePubMedGoogle Scholar
- Hu X, Zhang Y, Zhou X, Xu B, Yang M, Wang M, Zhang C, Li J, Bai R, Xu W, et al: Simultaneously Typing Nine Serotypes of Enteroviruses Associated with Hand, Foot, and Mouth Disease by a GeXP Analyzer-Based Multiplex Reverse Transcription-PCR Assay. J Clin Microbiol. 2012, 50 (2): 288-293. 10.1128/JCM.05828-11.View ArticlePubMedPubMed CentralGoogle Scholar
- Peiris JS, Tang WH, Chan KH, Khong PL, Guan Y, Lau YL, Chiu SS: Children with respiratory disease associated with metapneumovirus in Hong Kong. Emerg Infect Dis. 2003, 9 (6): 628-633. 10.3201/eid0906.030009.View ArticlePubMedPubMed CentralGoogle Scholar
- Allard A, Albinsson B, Wadell G: Detection of adenoviruses in stools from healthy persons and patients with diarrhea by two-step polymerase chain reaction. J Med Virol. 1992, 37 (2): 149-157.View ArticlePubMedGoogle Scholar
- Chung JY, Han TH, Kim CK, Kim SW: Bocavirus infection in hospitalized children, South Korea. Emerg Infect Dis. 2006, 12 (8): 1254-1256. 10.3201/eid1708.060261.View ArticlePubMedPubMed CentralGoogle Scholar
- Tabone T, Mather DE, Hayden MJ: Temperature switch PCR (TSP): Robust assay design for reliable amplification and genotyping of SNPs. BMC Genomics. 2009, 10: 580-10.1186/1471-2164-10-580.View ArticlePubMedPubMed CentralGoogle Scholar
- Raymond F, Carbonneau J, Boucher N, Robitaille L, Boisvert S, Wu WK, De Serres G, Boivin G, Corbeil J: Comparison of automated microarray detection with real-time PCR assays for detection of respiratory viruses in specimens obtained from children. J Clin Microbiol. 2009, 47 (3): 743-750. 10.1128/JCM.01297-08.View ArticlePubMedPubMed CentralGoogle Scholar
- Chen Y, Cui D, Zheng S, Yang S, Tong J, Yang D, Fan J, Zhang J, Lou B, Li X, et al: Simultaneous detection of influenza A, influenza B, and respiratory syncytial viruses and subtyping of influenza A H3N2 virus and H1N1 (2009) virus by multiplex real-time PCR. J Clin Microbiol. 2011, 49 (4): 1653-1656. 10.1128/JCM.02184-10.View ArticlePubMedPubMed CentralGoogle Scholar
- Hindiyeh M, Hillyard DR, Carroll KC: Evaluation of the Prodesse Hexaplex multiplex PCR assay for direct detection of seven respiratory viruses in clinical specimens. Am J Clin Pathol. 2001, 116 (2): 218-224. 10.1309/F1R7-XD6T-RN09-1U6L.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2334/12/189/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.