This article has Open Peer Review reports available.
Rapid and accurate detection of RMP- and INH- resistant Mycobacterium tuberculosisin spinal tuberculosis specimens by CapitalBio™ DNA microarray: A prospective validation study
- Zehua Zhang†1,
- Litao Li†2,
- Fei Luo1,
- Peng Cheng1,
- Feng Wu3,
- Zheng Wu1,
- Tianyong Hou1,
- Min Zhong4 and
- Jianzhong Xu1Email author
© Zehua et al.; licensee BioMed Central Ltd. 2012
Received: 25 March 2012
Accepted: 4 November 2012
Published: 14 November 2012
DNA microarrays can detect tuberculosis and its multi-drug resistant form in M. tuberculosis isolates and sputum specimens with high sensitivity and specificity. However, no performance data currently exists for its use in spinal tuberculosis specimens. This study was aimed to assess the performance of the CapitalBio™ DNA microarray in the detection of isoniazid (INH) and rifampicin (RMP) resistance in spinal tuberculosis compared with the BACT/MGIT 960 system.
From March 2009 to December 2011, 153 consecutive patients from Southwest Hospital, Chongqing with clinically and pathologically diagnosed spinal tuberculosis were enrolled into this study. Specimens collected during surgery from the tuberculosis patients were subjected to M. tuberculosis species identification and drug-resistance detection by the CapitalBio™ DNA microarray, and results were compared with those obtained from the absolute concentration drug susceptibility testing.
The CapitalBio™ DNA microarray achieved 93.55% sensitivity for the correct M. tuberculosis species identification of the 93 specimens that tested positive for spinal tuberculosis through culture. In addition, twenty-seven additional patients (45.0%) were detected by the DNA microarray to be positive for M. tuberculosis among sixty spinal tuberculosis patients who were culture negative. Moreover, the DNA microarray had a sensitivity of 88.9% and a specificity of 90.7% for RMP resistance, and the microarray had a sensitivity of 80.0% and a specificity of 91.0% for INH resistance. The mean turn-around time of M. tuberculosis species identification and drug resistance detection using the DNA microarray was 5.8 (range, 4–9) hours.
The CapitalBio™ DNA microarray is a feasible and accurate tool for the species identification of M. tuberculosis and for directly detecting RMP and INH resistance from spinal tuberculosis specimens in fewer than 9 hours.
Multi-drug resistant tuberculosis (MDR-TB) is becoming a serious public-health concern in developing countries. The World Health Organization (WHO) has estimated that there are 440,000 incident MDR-TB cases, and 150,000 persons with MDR-TB died worldwide in 2008 . China is currently ranked 1st among the 27 countries with the highest burden of MDR-TB and extensively drug-resistant tuberculosis (XDR-TB) in the world . As the most common extrapulmonary tuberculosis, spinal tuberculosis can lead to severe complications and is also facing the problem of drug resistance. The emergence of drug-resistant spinal tuberculosis may lead to an increased number of relapses and initial treatment failures . Therefore, early diagnosis of drug resistance is essential to the optimal management of drug-resistant spinal tuberculosis.
Conventional drug susceptibility testing (DST) can provide definitive results, but this approach is time-consuming and usually requires several weeks to produce susceptibility profiles, can lead to inadequate treatment and further acquired resistance during this period . Moreover, DST for tuberculosis spondylitis is not performed routinely in most resource-poor hospitals in China because of the biosafety concerns and inadequate infrastructure, which present a major hindrance for the treatment of the disease. Therefore, there is an urgent need to develop accurate, rapid and feasible molecular methods to detect the drug-resistance.
Recently, a multitude of commercial molecular DST kits, such as INNO-LiPA, Genotype MDR-TBplus, CapitalBio™ DNA microarray and Xpert MTB/RIF, have been developed for the detection of mutations associated with resistance to RMP and INH [4–12]. Among them, the CapitalBio™ DNA microarray that incorporate specific nucleotides at a given position of the rpoB, inhA and katG gene has been developed by CapitalBio Corporation to detect tuberculosis and its multi-drug resistant form in M. tuberculosis isolates and sputum specimens with notable sensitivity and specificity [6, 11, 12]. Previous study reported that the system had an accuracy of 91.8% in RMP susceptibility prediction and 70.2% in INH, compared with phenotypic DST . However, there is no data to support this technique’s application for testing spinal tuberculosis specimens. In our study, we aimed to access the feasibility and accuracy of the CapitalBio™ DNA microarray for parallel species identification and detection of mutations that confer INH and RMP resistance among consecutive spinal tuberculosis patients.
Materials and methods
From March 2009 through December 2011, we performed this prospective validation study in Southwest Hospital, Chongqing, China. One hundred and fifty-three consecutive patients with clinically and pathologically diagnosed spinal tuberculosis were enrolled into this study. Clinical specimens (e.g., pus, caseous or granulation tissue) were collected during surgery and were divided into two aliquots for tuberculosis detection; one of the aliquots was subjected to species identification and detection of gene mutations by the CapitalBio™ DNA microarray (CapitalBio Corp., Beijing, China), and the other aliquot was processed for culture and DST. The study was approved by the Ethics Committee of Southwest Hospital. Written informed consent was obtained from all of the patients.
Diagnostic criteria for the spinal tuberculosis
Diagnosis of spinal tuberculosis was made with reference to clinical and radiological findings and was verified histopathologically after debridement in all patients. Clinical and laboratorial presentation: Fever, night sweat and weight loss, back pain, neurological signs, angular deformity, positive response to anti-tuberculosis treatment, relative lymphocytosis, a low level of haemoglobin and a raised ESR. Radiology: X-ray, CT and MRI revealed bone destruction, vertebral collapse, kyphosis, tubercular lesions and paravertebral abscess. Histopathology: Epitheloid granulomas, a granular necrotic background and lymphocytic infiltration.
CapitalBio™ DNA microarray design and preparation
Assay Procedure for the CapitalBio™ DNA microarray
Spinal tuberculosis specimens were subjected to DNA microarray analysis at the Laboratory Department of Southwest Hospital. Investigators and laboratory staff were blinded to the culture and DST results of the clinical specimens. The CapitalBio™ DNA microarray was performed according to the manufacturer’s instructions [11, 12]. Briefly, 1–2 ml of the spinal tuberculosis specimen was inactivated for 30 min at 95°C followed by sonication in an ultrasonic water bath for 25 min. Next, the inactivated specimen was liquefied with an equal volume of liquefying solution (containing 0.5% N-acetylcysteine, 1.45% trisodium citrate and 2% NaOH) for 30 min at 37°C. After inactivation and liquefaction, 1 ml of the specimen was centrifuged at 12000 rpm for 5 min to pellet the bacteria. Following centrifugation, the supernatant was discarded, and the pellet was resuspended in 1 ml of 0.9% (w/v) saline and centrifuged at 12000 rpm for 5 min. This supernatant was subsequently discarded, and the pellet was resuspended in 50 μl of 10 mM Tris-EDTA buffer and then transferred to an extraction tube. The total DNA was isolated from the sample by vortexing the tube at maximum speed in an Extractor™ 36 (CapitalBio) for 5 min. Next, the extraction tube was incubated at 95°C for 5 min, centrifuged briefly and the supernatant was stored at −20°C until use. Next, asymmetric PCR was performed on the samples by a Peltier PTC225 thermal cycler (MJ Research, Watertown, MA) for two amplification rounds according to the manufacturer’s instructions. Following PCR amplification, chip hybridization was performed on the samples in a three-dimensional tilting agitator BioMixer II hybridization; an automated SlideWasher-8 (both from CapitalBio) was then used to wash and dry the hybridized slides. The fluorescent signal on the slides that was emitted by the microarrays was detected using a GeneArray scanner (LuxScan™-10K confocal laser scanner, CapitalBio, China), and the fluorescence intensities were quantified by the tailor-made software developed by CapitalBio. The whole protocol took less than 9 h to perform, with up to 4 specimens being analyzed for M. tuberculosis identification and 2 specimens being analyzed for drug resistance detection in parallel.
Conventional culture and DST
The quality-assured culture and DST was conducted using the BACT/MGIT 960 system and the absolute concentration method on L-J medium at the Department of Clinical Laboratory, Infectious Disease Medical Center in Chongqing. The specimens were processed according to standard methodologies . The following critical concentrations were used respectively: 1 and 10 μg/ml for INH, and 50 and 250 μg/ml for RMP.
We enrolled a total of 153 patients with spinal tuberculosis at Southwest Hospital during the study period. Among the 153 patients, 76 were female and 77 were male. The mean age was 36.4 years (range, 4 to 76 years). A total of 77.1% (118/153) of the patients were initial treatment cases, and the remaining 22.9% (35/153) were retreatment cases who had received previous chemotherapy for a mean period of 7.1 months (range, 3 to 74 months) at the time of enrollment. None of the patients was HIV-seropositive.
The results of conventional culture and DST
Among the 153 specimens that were cultured, a total of 60.1% of the specimens (93/153) produced a positive culture. Following culture testing, the positive specimens were subjected to DST. The results of the DST using the absolute concentration method on L-J medium showed that a total of 15/93 (16.13%) isolates were resistant to INH, 18/93 (19.35%) were resistant to RMP, and 11/93 (11.83%) of these isolates were resistant to both INH and RMP (MDR-TB). The mean turnaround time of culture and DST was 56.8 days (range, 49 to 77 days).
Strain identification of M. tuberculosisusing the DNA microarray
Overall sensitivity of the DNA microarray for the M. tuberculosis strains identification
Smear-Positive and Culture-Positive
Smear-Negative and Culture-Positive
Correct-no./total no. (%)
Detection of the RMP and INH resistance using the DNA microarray
Detection of M. tuberculosis rpoBmutants
Performance of the DNA microarray assay for detection of resistance to INH and RMP compared with phenotypic DST
DNA microarray assay
Correct-no./total no. (%)
Correct-no./total no. (%)
Detection of M. tuberculosis inhA and katGmutants
Of the 15 INH-resistant specimens, 8 specimens (53.3%) had a mutation in katG (codon 315), which is the most common mutation found in INH-resistant strains. Three of the specimens (20.0%) had a mutation in the inhA gene, C15T. Additionally, two low-level INH-resistant specimens (13.3%) displayed the wild-type hybridization pattern on the DNA microarray, and 2 of the specimens (13.3%) failed to produce the hybridization pattern. Part of the hybridization pattern of the INH-resistant isolates was shown in Figure 3d, e. Moreover, three of the 78 INH-susceptible samples (3.9%) showed a mutation at nucleotide 15 within the promoter of the inhA, and 4 of the 78 INH-susceptible samples (5.1%) failed to produce and results. The sensitivity and specificity of the INH resistance detection by the DNA microarray were 80.0% and 91.0%, respectively (Table 2). The mean turnaround time of M. tuberculosis identification and drug-resistance detection using DNA microarrays was 5.8 hours (range, 4 to 9 hours).
To respond to the urgent need for rapid and accurate detection of M. tuberculosis and drug resistance, a DNA microarray system was developed by the CapitalBio Corporation. This microarray can simultaneously identify 17 mycobacterial species by targeting the species-specific sequences in the 16S rRNA. Moreover, the specialized microarray consists of 16 oligonucleotide probes, which allowed the detection of the 13 most common mutations in the rpoB gene, 2 mutations in the katG gene, and one mutation in the promoter region of the inhA gene. In our study, the identification of M. tuberculosis strains and detection of their resistant form in spinal tuberculosis specimens were evaluated by the CapitalBio™ DNA microarray assay.
We evaluated the utility of the CapitalBio™ DNA microarray in 153 consecutive spinal tuberculosis samples. The microarray system identified more than 93% of all cases with culture-confirmed spinal tuberculosis, including more than 73% of patients with smear-negative cultures. Moreover, this assay was positive in 45% (27/60) of the patients who were not confirmed by M. tuberculosis culture but were demonstrated both clinically and pathologically to have spinal tuberculosis. Results of the study revealed that the microarray assay has a higher susceptibility compared with the conventional BACT/MGIT 960 system for the detection of M. tuberculosis strains. Previous studies conducted by Guo and Zhu et al. also reported that for the direct detection of M. tuberculosis complex bacteria the overall sensitivity obtained with this commercial DNA microarray were 100% [11, 12]. The negative control of the species identification has not been setup considering the following reasons: First, Guo and Zhu respectively found that the specificity of the CapitalBio™ DNA microarray was 100% for the species identification of M. tuberculosis[11, 12]. Second, recruitment and enrollment of pyogenic spondylitis patients, with similar clinical characteristics and clinical specimens to spinal tuberculosis, are relatively difficult in our department. However, absence of the negative control might be a potential limitation of this study.
For the detection of anti-tuberculosis drug resistance, the DNA microarray had a sensitivity of 73.3% and specificity of 91.0% for INH resistance, and a sensitivity of 88.9% and specificity of 90.7% for RMP resistance. Interestingly, 5.6% of phenotypically defined RMP-resistant specimens and 13.3% of INH- resistant specimens had no mutations in related genes, suggesting that other mechanisms or mutations in other codons of the related genes may be responsible for the emergence of RMP and INH resistance. Moreover, a small proportion of specimens failed to produce results; one possible reason for the lack of results may be because of the exiguous M. tuberculosis bacilli contained in some specimens, or the DNA amplification inhibitors in the clinical specimens, resulting in a lack of DNA being extracted from these samples. Results of the DNA microarray assay could be obtained within a mean turnaround time of 5.8 hours from the start of the analysis.
Recently, some novel molecular DST methods based on nucleic acid amplification have been available. Among them, the Xpert MTB/RIF assay was endorsed by the WHO as a replacement for sputum smear microscopy [7, 8, 14–16]. Compared with the Xpert MTB/RIF assay, the CapitalBio™ microarray system has some limitations, e.g., the assay is semi-automatic, requires more experienced technical skills for manual manipulation and has more opportunities to create biohazard intermediates. However, the microarray system has two advantages compared to the Xpert MTB/RIF assays. First, the microarray assay is less costly because cartridges of the Xpert MTB/RIF are disposable. Second, the microarray assay can detect the gene mutations associated to both RMP and INH.
In addition, compared with isolates and sputum samples, the application of the DNA microarray in spinal tuberculosis samples is far more complex because of the diversity of clinical sample types, difficulties in obtaining adequate tissue for analyses and the methods of processing samples prior to analysis .
Overall, the CapitalBio™ DNA microarray is a simple and accurate tool for M. tuberculosis identification and for the diagnosis of RMP and INH resistance, demonstrating that the system is likely to be applicable to spinal tuberculosis specimens. The automated readout and short turnaround time make this assay suitable for testing and decreases delays in diagnosis, without the need to build large numbers of advanced biosafety facilities. Nevertheless, the infrastructure and trained personnel required for this system are still not available except in a limited number of reference laboratories, which dramatically reduces its clinical utility in poverty-stricken zones. Thus, the integral and fully automatic DNA microarray system should be developed, which would have direct benefits to workflows and to biosafety in resource-scarce settings. The gene mutations that relate to other first-line and second-line drug resistance should be considered for incorporation into the further DNA microarray system because of the dissemination of second-line drug-resistant M. tuberculosis strains. Additional large-scale studies are needed to more further evaluate its diagnostic performance in spinal tuberculosis.
We acknowledge the data collectors and the study participants from the Laboratory Department of Southwest Hospital. We also give our appreciation to the study participants from the Department of Clinical Laboratory of Infectious Disease Medical Center of Chongqing for their kind assistance during drug resistance testing.
- World Health Organization: Global tuberculosis control 2010. 2010, Geneva: WHO, IOP Publishing PhysicsWeb. http://www.who.int/tb/publications/global_report/en/ Google Scholar
- Li L, Zhang Z, Jianzhong X: Management of drug-resistant spinal tuberculosis with a combination of surgery and individualized chemotherapy: a retrospective analysis of thirty-five patients. Int Orthop. 2011, 36: 277-283.View ArticlePubMedPubMed CentralGoogle Scholar
- Rüsch S, Gaby E, Casal M: Multicenter laboratory validation of the BACTEC MGIT 960 technique for testing susceptibilities of mycobacterium tuberculosis to classical second-line drugs and newer antimicrobials. J Clin Microbiol. 2006, 44 (3): 688-692. 10.1128/JCM.44.3.688-692.2006.View ArticleGoogle Scholar
- Sun A, Fan X, An W, Yan J: Rapid Detection of rpoB Gene Mutations in Rif-resistant M.tuberculosis Isolates by Oligonucleotide Microarray. Biomed Environ Sci. 2009, 22: 253-258. 10.1016/S0895-3988(09)60053-2.View ArticlePubMedGoogle Scholar
- Cabibbe AM, Miotto P, Lazzeri E, Mugasa J, Santoro F: New DNA microarray platform for detection of MDR Mycobacterium tuberculosis and of drug-resistant malaria. Clin Microbiol Infect. 2011, 17 (Suppl 3): 591-592.Google Scholar
- Yao C, Zhu T, Li Y, Zhang L, Zhang B, Huang J, Fu W: Detection of rpoB, katG and inhA gene mutations in Mycobacterium tuberculosis clinical isolates from Chongqing as determined by microarray. Clin Microbiol Infect. 2010, 16 (11): 1639-1643. 10.1111/j.1469-0691.2010.03267.x.View ArticlePubMedGoogle Scholar
- Catharina C, Boehme CC, Pamela N: Rapid molecular detection of tuberculosis and rifampin resistance. N Engl J Med. 2010, 363 (11): 1005-1015. 10.1056/NEJMoa0907847.View ArticleGoogle Scholar
- Boehme CC, Nicol MP, Nabeta P: Feasibility, diagnostic accuracy, and effectiveness of decentralised use of the Xpert MTB/RIF test for diagnosis of tuberculosis and multidrug resistance: a multicentre implementation study. Lancet. 2011, 377 (9776): 1495-1505. 10.1016/S0140-6736(11)60438-8.View ArticlePubMedPubMed CentralGoogle Scholar
- Bazira J, Asiimwe BB, Joloba ML: Use of the GenoType® MTBDRplus assay to assess drug resistance of Mycobacterium tuberculosis isolates from patients in rural Uganda. BMC Clin Pathol. 2010, 10: 5-10.1186/1472-6890-10-5.View ArticlePubMedPubMed CentralGoogle Scholar
- Abebe G, Paasch F, Apers L: Tuberculosis drug resistance testing by molecular methods: Opportunities and challenges in resource limited settings. Journal of Microbiological Methods. 2011, 84: 155-160. 10.1016/j.mimet.2010.11.014.View ArticlePubMedGoogle Scholar
- Guo Y, Zhou Y, Wang C, Cheng J: Rapid, accurate determination of multidrug resistance in M. tuberculosis isolates and sputum using a biochip system. Int J Tuberc Lung Dis. 2009, 13 (7): 914-920.PubMedGoogle Scholar
- Zhu L, Jiang G, Wang S, Guo Y: Biochip system for rapid and accurate identification of Mycobacterial species from isolates and sputum. J Clin Microbiol. 2010, 48 (10): 3654-3660. 10.1128/JCM.00158-10.View ArticlePubMedPubMed CentralGoogle Scholar
- The basis speciality committee on bas science of Chinese Anti-tuberculosis Association: The laboratory science procedure of diagnostic bacteriology in tuberculosis. Chinese Journal of Anti-tuberculosis. 1996, 18 (1): 28-31.Google Scholar
- Lawn SD, Zumla AI: Diagnosis of extrapulmonary tuberculosis using the Xpert MTB/RIF assay. Expert Rev Anti Infect Ther. 2012, 10 (6): 631-635. 10.1586/eri.12.43.View ArticlePubMedPubMed CentralGoogle Scholar
- Suzana S, Shalini B, Rupali P, et al: Rapid diagnosis of extra pulmonary tuberculosis by automated Xpert MTB/RIF assay. BMC Infect Dis. 2012, 12 (Suppl 1): P32-View ArticlePubMed CentralGoogle Scholar
- Vadwai V, Boehme C, Nabeta P, et al: Xpert MTB/RIF: a new pillar in diagnosis of extrapulmonary tuberculosis. J Clin Microbiol. 2011, 49 (7): 2540-2545. 10.1128/JCM.02319-10.View ArticlePubMedPubMed CentralGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2334/12/303/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.